ACCURACY: ± 1.5% Full Scale
DIAL SIZE: 3”, 4” or 5”
DIAL MATERIAL: White background with black marks. Anodized Aluminum background available.
STEM LENGTH: 2” to 36”
STEM DIAMETER: 1/4”
HEAD, BEZEL, MOUNTING BUSHING, STEMS: 304SS
OPERATING CONDITIONS: Head temperature should not exceed 200°F (150°F if silicone filled). Stem should not be exposed to continuous temperatures exceeding 50% over-range or 800°F(550°F if silicone filled).
TEMPERATURE RANGES: See next page for complete list of ranges.
CASE RATING: IP65, Hermetically Sealed
LENS: Glass (std), Acrylic, Polycarbonate, Tempered Glass, Laminated Safety Glass
IMMERSION: Minimum 2” in liquid, and 4” in gas for most ranges. Certain ranges require up to 4” in liquids & 5” in gas.
MOUNTING CONNECTION: 1/2” NPT (std), 1/2” BSPT, 1/4” NPT, 1/4” BSPP
TEMPERATURE SENSING AREA: Last 2” to 4” of the stem
NOTE: Thermowells should be used whenever the stem or bulb would be exposed to pressure, corrosion, velocity, abrasion or shear forces. Thermowells also make it possible to remove the thermometer without losing pressure or the contents of the process.
Download Install Instructions: